Spare part number: 800-901010
The SensorInterface2 board is collecting the preamp signals with its 16-bit AD converter and is doing some signal filtering on it. With the high precision reference and the crystal based frequency measurement, it keeps the electronic part of the measuring stable. Because this board is used on different places in the airpointer it is very important to have the jumper set and the address switch set correctly. Jumper List It is mainly used together with Thermo_i benches. On some setups we use a I2C interface to communicate with other boards, connecting or disconnecting cables with power on gives you a 50:50 chance to destroy the boards! Always power off the unit!